The Chroma A W and A W DC electronic load mainframes accept the user-installable A series load modules for easy system. RENT CHROMA DC Electronic Load Mainframe. Chroma Model DC Electronic Load Mainframe is suitable for the test and evaluation of multi-output. Product Name: DC Electronic Load. Model Name: , , , , , , , , , Manufacturer’s Name: Chroma ATE .
|Published (Last):||23 June 2017|
|PDF File Size:||10.18 Mb|
|ePub File Size:||18.27 Mb|
|Price:||Free* [*Free Regsitration Required]|
Mouse move on Image to zoom. The holds four load modules at most to offer 8-channel W input load with standard front-panel inputs. What can i do to improve my program?
The loads can be operated in constant current, constant voltage, constant power and constant resistance and may be placed in parallel for increased current and power. Some instruments allow part or all of the lowercase command to be omitted, others allow only for the whole part to be present or omitted but not fractions thereof. Figure 7 shows the ripple current waveform from a LED load. The steps i have to do: Its just that ive been using the wrong commands from the manual so my electronic load wouldnt accept it.
The built-in resistive load current sensing circuit is capable of measuring current with an accuracy of 0.
Chroma 6314 Programmable DC Electronic loads
Standard accessories supplied by the original manufacturer No manuals or accessories required Other specify Will you accept an instrument with minor cosmetic and physical flaws if it operates properly? Additionally, the A series offers an optional remote controller for automated production lines.
Thank you for reminding me about the upper case and lower case. Message 4 of 7. Please let us know which of these you would be willing to consider for the price discount.
Two or more load modules can be paralleled together to achieve the desired loading. Yes No Are you 63114 to prepay your order by check or wire transfer to avoid credit card fees?
Please provide the requested information to receive an estimated shipping cost. Additional Info Please let us know any information that will help us fill your requirement such as urgency, quantity needed, etc.
So when i simulate a voltage into the load through a power source, the load display will only receive till 10V no matter if i tune the source to above 10V. The load can be operated in constant current, constant voltage, and constant resistance.
OCP Test By simply choosing the channel and setting the OCP parameters start current, end current, step current and dwell time from the front panel, the A series provides a fast cnroma easy OCP testing solution. Please correct the issue indicated and resubmit or contact us via email or toll free at and we will be happy to assist you with this request.
Chroma 6314 DC Electronic Load Mainframe With Manual O1
Add to Inquiry Cart Data Sheet. Unable to Process Request Your offer was received but we are unable to process it for the following reason: Figure 10 shows the current waveform from a CR mode of an Electronic Load loading. For cheoma, the test of a battery charger can be simulated easily by setting the load to operate in constant voltage.
The A design also has increased bandwidth to chdoma for PWM dimming testing. This makes it inconvenient for mass production testing. Message 1 of 7. Message 6 of 7. These are the commands given in the manual book.
Chroma DC Electronic Load Mainframe With Manual O1 | eBay
The waveforms programmable parameters include: The command line syntax for those instruments is usually case insensitive and the uppercase part of the command is mandatory while the lower case part can be ommitted but doesn’t have to.
Use data flow instead! I have 4 double modules. The system is configured by plugging the user selectable load modules into the system mainframe, and operated using the keypad on the front panel of the instrument or the remote controlled instructions via RS or GPIB interface.
Error Sending Request We are very sorry but an error on the server is preventing your request from being processed. Figure 5 shows the dimming current waveform when using Chromz as a load. I purposely put both cases so it would be easier for me. Contact Info Please let us know how we can contact you.
DC Electronic Load | Chroma ATE Inc.
The A series has a self-diagnosis routines to maintain instrument performance. When using a resistor as loading, the V-I curve of the resistor is not able to chrona the V-I curve of the LED as shown in blue on figure 1.
Message 5 of 7. Each model is designed with specific applications in mind.